Effect of cough technique and cryogen gas on temperatures achieved during simulated cryotherapy
© Seamans et al; licensee BioMed Central Ltd. 2007
Received: 07 April 2007
Accepted: 01 October 2007
Published: 01 October 2007
Cryotherapy is often used to treat cervical precancer in developing countries. There are different methods and cryogen gases used for cryotherapy, including the freeze-flush-freeze (cough) technique employed to minimize gas blockage. However, there is limited information to compare their effectiveness.
Using a tissue model, we compared temperature-time curves for four cryotherapy methods: uninterrupted freezing with nitrous oxide (N2O) and carbon dioxide (CO2), and two methods using a standard and extended version of the cough technique with CO2.
Uninterrupted freezing with both N2O and CO2 produced tissue temperatures less than -20°C (-40°C and -30°C respectively). CO2 cryotherapy procedures using the two cough techniques produced temperatures greater than -20°C in the model tissue.
CO2 cryotherapy using the cough technique may not achieve sufficiently low temperatures to produce the desired therapeutic effect. Other alternatives to the prevention of gas blockage should be developed.
Cryotherapy is a method for the treatment of cervical precancerous lesions [1, 2] that is considered the most suitable option to use in low-resource settings with underserved populations because it is low cost, requires no anaesthesia or electricity, and has a low complication rate . Although nitrous oxide (N2O)-based cryotherapy is effective [4, 5], its limited availability and higher cost in developing countries make carbon dioxide (CO2)-based cryotherapy a more practical choice. The reported limitations to using CO2 include blockage of the cryogun during the cryotherapy procedure, inherently warmer tip temperatures due to the physical properties of the gas, and indications of lower therapeutic effect due to insufficient depth of tissue destruction during studies that employed a freeze-flush-freeze cycle ("cough technique") .
The "cough technique" has been recommended to avoid cryogun blockage during treatment . It involves a cyclic momentary thawing of the cryogun probe by systematically interrupting freezing. Although the cough technique is currently being used in developing-country low-resource settings, there are no published data on the validity of this approach or the impact of using the cough technique on the depth of tissue necrosis generated by these methods. We developed a system to compare the temperature profile at the probe and in a tissue model of three different CO2-based methods (an uninterrupted freeze cycle, the standard cough technique, and an extended cough technique) as well as the well-characterized N2O uninterrupted freeze.
We evaluated three CO2-based cryotherapy techniques: uninterrupted freezing, the standard cough technique, and an extended cough technique. For the standard cough technique, the operator depressed the defrost key every 15 to 20 seconds for approximately one second, released it, and immediately reengaged the freeze key. To employ the extended cough technique, the operator inserted a short pause (approximately onesecond) after releasing the defrost key and before reengaging the freeze key. During this pause, the operator waited to hear the cryogun pulse valve open before reengaging the freeze key.
Individual measurements for each method were reproduced five times. Uninterrupted freezing using N2O cryotherapy was our referent standard.
The five runs for each group were averaged for a mean curve. The Kruskal-Wallis test (for nonparametric data) was used to test for statistical differences (P < 0.05) between median temperatures for the mean curve for each cryotherapy technique.
A summary of the temperature-time curves from 10 to 180 seconds
CO2 (Extended Cough)
CO2 (Standard Cough)
Time to -20°C (sec)*
Time to -20°C (sec)*
This laboratory-based evidence indicates that both cough techniques, used to prevent blockage of the cryogun during CO2 cryotherapy, significantly affected the temperature achieved at a 5 mm depth and did not achieve the minimum temperature of -20°C required for tissue destruction and therapeutic effects . When compared to the temperatures achieved with N2O cryotherapy, the referent treatment shown to be effective for treatment of cervical precancer, it suggests that neither cough technique is likely to provide the same therapeutic effect as uninterrupted freeze. In addition to the importance of minimum temperature required for necrosis, there are other factors such as cooling rate and total time at the minimum temperature  that are also affected by the cough technique.
The standard cough technique was limited in the minimum temperature achieved due to the design of the Wallach cryogun internal pulse valve. The valve must be closed during the defrost cycle and open during the freeze cycle. The rapid cycling of the standard cough technique does not allow time for the valve to reopen following defrost. It remains closed causing the cryogun to function at an intermediate temperature between defrost and freeze. The extended cough (increased time interval between the freeze cycles) allows time for the valve to open, ensuring proper functioning during the subsequent freeze cycle. However, both methods caused significant temperature variation at the cryogun probe and limit the minimum temperature achieved in the tissue.
We acknowledge that this is a model system in which temperature was measured at a specific depth rather than the tissue necrosis, the true desired outcome. Additionally, because the study employed model tissue rather than human tissue, additional human studies will be needed to confirm our laboratory findings. Thus, we can only make qualitative assessments between different methods. While our ability to extrapolate as to what will happen in live tissue is limited, it can be anticipated that these higher than desirable temperatures will be further exacerbated because of the warming effect of the blood supply leading to higher tissue temperatures than observed in our model system.
This evaluation was limited to use of the Wallach LL100; minimum probe and tissue temperatures may vary if other cryoguns are used. In addition, frequency of device blockage is undetermined when using CO2 with other cryotherapy devices, which may not incorporate an "active" defrost feature and hence the cough technique may or may not be possible.
Based on these data, technical improvements for CO2 cryotherapy are necessary to make it a more useful treatment modality for cervical precancer in underserved populations who cannot get surgical excision. This may include methods to dry or condition CO2 to prevent the blockage in the cryogun during the procedure.
This research was supported by the Intramural Research Program of the U.S. National Institutes of Health and National Cancer Institute.
- Gaffikin L, Blumenthal PD, Emerson M, Limpaphayom K: Safety, acceptability, and feasibility of a single-visit approach to cervical-cancer prevention in rural Thailand: a demonstration project. Lancet. 2003, 361: 814-820. 10.1016/S0140-6736(03)13713-0.View ArticlePubMedGoogle Scholar
- Adewole IF, Babarinsa IA, Odeniyi GD: Cryotherapy in the management of cervical intraepithelial neoplasia in developing countries. Int J Gynaecol Obstet. 1998, 60: 69-70. 10.1016/S0020-7292(97)00222-1.View ArticlePubMedGoogle Scholar
- Jacob M, Broekhuizen FF, Castro W, Sellors J: Experience using cryotherapy for treatment of cervical precancerous lesions in low-resource settings. Int J Gynaecol Obstet. 2005, 89 Suppl 2: S13-S20. 10.1016/j.ijgo.2005.01.026.View ArticlePubMedGoogle Scholar
- Mitchell MF, Tortolero-Luna G, Cook E, Whittaker L, Rhodes-Morris H, Silva E: A randomized clinical trial of cryotherapy, laser vaporization, and loop electrosurgical excision for treatment of squamous intraepithelial lesions of the cervix. Obstet Gynecol. 1998, 92: 737-744. 10.1016/S0029-7844(98)00246-4.View ArticlePubMedGoogle Scholar
- Alliance for Cervical Cancer Prevention (ACCP): Effectiveness, Safety, and Acceptability of Cryotherapy, a systematic literature review. 2003Google Scholar
- Santos CL, Torres J, Sanchez J, Dasgupta A, Jeronimo J: Lack of effectiveness of CO2 cryotherapy for treatment of CIN. Int J Gynaecol Obstet. 2004, 44-45. 10.1016/j.ijgo.2004.06.014.Google Scholar
- Gage AA: What temperature is lethal for cells?. J Dermatol Surg Oncol. 1979, 5: 459-60, 464.View ArticlePubMedGoogle Scholar
- Hoffmann NE, Bischof JC: The cryobiology of cryosurgical injury. Urology. 2002, 60: 40-49. 10.1016/S0090-4295(02)01683-7.View ArticlePubMedGoogle Scholar
- Seamans Y, Sellors J, Broekhuizen F, Howard M: Preliminary report of a gas conditioner to improve operational reliability of cryotherapy in developing countries. BMC Womens Health. 2006, 6:2.: 2-10.1186/1472-6874-6-2.View ArticleGoogle Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1472-6874/7/16/prepub
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.